Sloan Bedpan Washer After 1998 with Deoseptic Unit Parts Breakdown and Troubleshooting Guide
Sloan Bedpan Washer Breakdown PDF
• Bedpan Flush Valve with Deoseptic
- 1. #08725 - V600AA - 1-1/2" x 21-1/2" Chrome Plated Vacuum Breaker
#80795 - V600AA - 1-1/2" x 13-1/2" Chrome Plated Vacuum Breaker (ADA Height)
#08436 - V600AA - 1-1/2" x 21-1/2" Chrome Plated Vacuum Breaker with 2" Offset
- 2. #08438 - V651ABX - Vacuum Breaker Repair Kit
- 3. #08812 - F2A - 1-1/2" Slip Joint Coupling
- 4. #08329 - DV1 - Diverter Valve Body (Non-Deoseptic Bottle)
#80820 - DV17 - Diverter Valve Body (Deoseptic Bottle)
- 5. #08813 - DV1001A - Wall Support Kit
- 6. #04898 - Equiparts - Outlet Tube Gasket
- 7a. #80821 - F1005 - Outlet Tube
7b. #08073 - F182 - Outlet Tube with 1-1/2" Offset
- 8. #80822 - F2 - 1-1/2" Coupling Nut
- 9. #08017 - F5AT - 1-1/2" Spud Coupling Assembly
- 10. #08093 - DV32AA - Spray Arm Replacement Kit
#87063 - DV1014A - Short Spray Arm Replacement Kit
- 11. #08846 - DV1007A - Spray Arm Repair Kit
- 12. #08803 - DV1003A - Pivot Replacement Kit
- 13. #08802 - DV1002A - Deoseptic Pivot Replacement Kit
13a. #08354 - DO6 - O-Ring Piston (2 Required)
13b. #08355 - DV4 - Set Screw
13c. #08361 - DV16 - Diverter Valve Seal Washer
13d. #08360 - DV14 - Diverter Valve Seal
- 14. #80823 - DV21 - Plug Button
- 15. #80824 - DV33A - Swivel & Ball Check Assembly
- 16. #80825 - DV36A - Aspirator Tube Assembly
- 17. #80826 - DV29 - Deoseptic Solution Bottle
- #08092 - DV50A - Repair Seal Set (Most Common)
Troubleshooting Q&ABelow is a Troubleshooting Q&A Guide for the Bedpan Washer Flushometers from Sloan. The guide highlights the most commonly occurring problems and how to go about fixing them. As always you can call us for assistance or if you need to replace a part and/or unit.
Diverter leaking at spray arm
A) Diverter spray arm (old style) is damaged.
B) Spray arm o-ring is damaged or installed improperly.
A) Replace diverter arm with new spray arm replacement kit #08093. NOTE: On old style diverter spray arm, the spray arm o-ring was placed on groove at the end of the spray arm. When installing new spray arm kit, place washer into bottom of hole of nozzle pivot. Then place the o-ring on top of washer. Insert new spray arm into the pivot, pressing in slightly. While holding spray arm in place, tighten set screw.
B) Re-Install o-ring as described above in step (A.) If leakage persists, replace using spray arm kit #08093.
Diverter leaking at nozzle pivot
A) One or both pivot o-rings are worn.
B) Diverter seal washer worn or missing.
C) Nozzle pivot damaged or defective.
A) Replace nozzle pivot assembly.
B) Replace pivot seals. Use diverter rebuild kit.
C) Replace nozzle pivot assembly. Use pivot replacement kit.
Erratic spray pattern from spray arm nozzle
A) Debris clogging nozzle screen.
B) Damaged or missing nozzle screen or spray disc.
A) Wash nozzle screen under running water.
B) Replace using #08846 spray arm repair kit.
Deoseptic solution NOT siphoning from bottle during flush cycle (Deoseptic models only.)
A) Adjusting screw not open.
B) Aspirator tube screen clogged.
C) Aspirator tube not installed completely onto swivel & ball check assembly.
D) Aspirator tube cracked.
E) Rubber ball check stuck inside swivel & ball check assembly.
A) Open adjusting screw until adequate amount of solution is siphoned from bottle during flush cycle.
B) Remove solution bottle. Remove aspirator tube from swivel & ball check assembly and wash screen under running water.
C) Remove solution bottle. Make sure aspirator tube is pushed completely onto nipple of swivel & ball check assembly.
D) Replace aspirator tube assembly.
E) Remove solution bottle and aspirator tube from swivel & ball check assembly. Gently insert small implement (paper clip, small dia. hex key, etc.) into nipple of swivel & ball check assembly to dislodge rubber ball check.
Water leaking back into solution bottle
A) Rubber ball damaged or missing.
B) Rubber ball not sitting in seat of swivel & ball check assembly.
A) Replace rubber ball using Deoseptic Diverter Rebuild Kit. Includes Outlet Tube Gasket (item 6,) Spray Arm Repair Kit (item 11,) and all items in Pivot Replacement Kit (item 12) except Pivot.
B) Remove swivel and ball check assembly, taking care not to lose rubber ball. Insert rubber ball back into seat of swivel & ball check assembly and reassemble. NOTE: Follow steps in solution (E) of problem 4.
Sloan Bedpan Washer TroubleShooting PDF
View the Sloan Bedpan Washer repair parts or our selection of Sloan flush valves.